|Method||High Performance Liquid Chromatography|
|Standards||ASTM D 5837, IEC 61125|
What is it used for?
The breakdown of the cellulose-based windings in the solid insulation of transformers give rise to furanic compounds, and therefore act as a good marker of aging or fault conditions.
How does it work?
A portion of the oil sample is collected, with furan compounds separated from the oil using solid phase extraction (SPE). These compounds are then collected and analysed by liquid chromatography using a UV detector.
What do the results tell us?
The concentration of each of the five taget furan compounds (fufural, furfural, hydroxymethyl-2-furaldehyde, 2-acetyl furan and 5-methyl-2-furaldehyde, all in terms of mg/L.
Back to Testing Methods